On World Metrology Day, May 20, 2019, new values were implemented for the International System of Units (SI) base units of the kilogram, kelvin, mole and ampere, and the derived units of the volt and ohm. These values are based on redefinitions of the Planck constant, elementary charge, the Boltzmann constant and the Avogadro constant. The changes are based on the evolution of the definition of the kilogram, the SI unit of mass, which has progressed from a physical artifact-held at the International Bureau of Weight and Measures (BIPM) since 1889 to a constant of nature, Planck’s constant, that can be accessed anywhere in the universe. The new definition of the kilogram benefits the entire world in potentially having better access to realization of the kilogram, and eliminates the risks associated with maintaining a standard that is based on a sole artifact.

As a result of the redefinitions, some measurement reference standards must be adjusted to the new value of the definition. Other measurement reference standards may account for the redefined value by incorporating an additional contributor to their measurement uncertainty evaluations. This application note summarizes the changes being made to SI units and provides guidance on how to implement them.

**SI units have a history of change**

This is not the first time that an
adjustment to the measurement system has been made. For example, in 1990 the
volt was defined to be based on the Josephson effect and the ohm on
the Quantum Hall effect. At that time, the shift^{1} of the
volt in the United States was 9.26 x 10-6 (9.26 parts per million or 0.000926%)^{1} which
was significant because a large percentage of laboratories owned devices that
were capable of this accuracy, such as the Fluke 5700A High
Performance Multifunction Calibrator. The 1990 volt definition exceeds the performance
specification of this product on the 11 volt and 22 volt ranges, so a Fluke
5700A that was calibrated to the United States representation of the volt in
1989 would be out of tolerance as compared to the 1990 volt. The Fluke 5700A
not only was used at many National Metrology Institutes (NMIs), but it was
used in primary, secondary, and working level calibration laboratories and in
some cases, test laboratories around the world, as the Fluke 5730A is
today.

The adjustments to the volt on May 20, 2019 is nearly 100 times less in magnitude than the 1990 adjustment, and as a result a smaller number of organizations will be affected. However, metrologists should be aware of the redefinition and its implications on their calibration laboratory.

**Experiments to transition to the new definition of the kilogram and the impact on other SI definitions**

In order to transition from the International Prototype kilogram (IPK) to a constant of nature, a series of experiments have been conducted in collaboration with the top NMIs in the world. The experiments involved realizing the kilogram through the Kibble (watt) balance and through means of x-ray-crystal-density. As a result of the experiments, the following constants will be updated on May 20, 2019:

- The Planck constant h is exactly 6.626 070 15 ×10
^{-34}joule second - The elementary charge e is exactly 1.602 176 634 ×10
^{-19}coulomb - The Boltzmann constant k is exactly 1.380 649 ×10
^{-23}joule per kelvin - The Avogadro constant NA is exactly
6.022 140 76 ×10
^{23}mol^{-1}

The SI will continue to have the same seven base units, but the new definition of the kilogram will now be expressed in terms of the Planck constant; the ampere will be defined in terms of the elementary charge; the kelvin will be defined in terms of the Boltzmann constant; and the mole will be defined in terms of the Avogadro constant.

**Measurement system adjustments as a result of constant redefinitions**

**The Volt V**

The most significant shift in any of the
measurement units is the volt. The updates to the Planck Constant and the
Fundamental Electron Charge will cause a shift^{3} of 0.107 x 10^{-6} or
0.107 parts per million (ppm). The new value for the Josephson constant (2e/h)
is slightly smaller than the 1990 value, so a device that is measured to the
2019 value will be larger than the 1990 value by approximately 0.1 ppm^{8}.
Calibration laboratories that operate a Josephson Voltage System
(JVS) to directly realize the volt will be required to update the value
for the Josephson constant in their system operating software. We advise these
laboratories to learn how to update the software before May 20, 2019,
but not complete the update until that time. The only other commercial
instrument that can observe this adjustment is the Fluke 732 series of
Direct Voltage Standards. When a Fluke 732 is calibrated against a JVS, the
uncertainty produced for the calibration is typically between 0.06 ppm to 0.1
ppm. While the adjusted value will equal the calibration uncertainty for the
732 series instruments, the specification component of long-term stability is
much larger than this, and even when maintaining a control chart that utilizes
linear regression, the value associated with this adjustment is absorbed in the
linear regression uncertainty^{4}.

The International Committee of Weights and Measures (CIPM) Consultative Committee for Electricity and Magnatism (CCEM) has produced a guideline which states that no action is required if your voltage related uncertainties are larger than 0.25 ppm^{5}. If your uncertainty for the volt is less than 0.25 ppm and you do not have direct access to a JVS, we recommend that you add the 0.107 ppm adjustment quantity to your uncertainty analysis.** **

**The Ohm ****Ω**

The adjustment to the ohm is much smaller than the volt, approximately 0.02 ppm^{3}. If your laboratory is operating a Quantum Hall based resistance standard (QHR), the value for the Von Klitzing Constant should be updated in the operating system software in the same manner as described for the volt. However, since most calibration laboratories do not operate a QHR, the value of this adjustment is generally insignificant as it is half the calibration uncertainty that NIST provides for a Thomas 1 ohm calibration^{6}. The guidance from the CCEM is that no action is required for resistance uncertainties that are larger than 0.05 ppm.** **

**The Kilogram kg**

On May 20, 2019, calibrations that are traceable to the IPK will have an increased expanded uncertainty of 0.02 ppm^{7}, which is less than half the uncertainty of a NIST calibration at 1 kg, so this adjustment is insignificant to industrial calibration laboratories.** **

**The Pascal Pa**

The fundamental SI unit of pressure, the
Pascal is expressed in N/m^{2} which can be further reduced to
kg/m•s^{2}. Since the meter and second are not being
redefined in 2019, the only change is based on the kilogram, which again is
approximately 0.02 ppm. The most accurate calibrations for pressure are on the
order of ppm, so again, this adjustment is insignificant.

** The Kelvin K**

Although the definition of the SI unit for thermodynamic temperature, the kelvin, is being redefined in terms of the Boltzmann constant, the definitions of ITS-90 are not changing at this time, so any calibration that is performed in accordance with ITS-90 or PLTS-2000 will not change.

Did you like the content? Did you take any doubts? Take advantage and check all the content and products available on the Datasonic website, a pioneer in cutting-edge technological equipment that stands out for its diversity of brands, models and especially prices is at your disposal, so you can take all your doubts, compare prices and finally, choose the best product for you. The Datasonic website has a wide portfolio and a range of varied products, all at your disposal. The Datasonic website is available to you 24 hours a day, 7 days a week. Just waiting for you to come and check out all the technology in equipment we provide for you and your company. What are you waiting for? Visit us right now.

Access: https://datasonic.com.br